Spectralis diagnostic equipment Integration with Electronic Health Record
Could not connect and archive to network storage.
This is an ophthalmic imaging platform. BeCloud has successfully integrated this equipment with various EHR software providers providing a seamless workflow for eye clinics.
Our client purchased a new Spectralis ophthalmic imaging platform. Because the imaging files can be huge and the new device had limited on board storage we recommended network attached storage. There are many options for providing this type of storage but we have had the most success with Synology network storage devices. In this case, the OCT machine could not connect to the network storage device. After much troubleshooting and contact with Heildeberg Engineering support BeCloud resolved the issue by enabling SAMBA 3 on the network storage device.